The Ontoto Connect Android & iOS application provides a gateway to communicate with the loggers & their data, providing customers with a fast and efficient way of initially installing and configuring their device & giving data insights at a glance. Click here to download the application for android. The iOS version can be installed here on the iOS App Store.
- Offline configuration is used when setting up the logger initially. Your mobile needs to be within a 10 m radius of the logger
- Online configuration is used when the logger is active. For Example, the reporting cycle of the logger can be configured remotely without going to the site. Our loggers always communicate with the Ontoto server via API to receive updates such as configuration changes and firmware updates.
Easily connect, configure & start your data logger via bluetooth with Ontoto connect.
Designed with simplicity, its easy to get your data logger up and running.
Ontoto connect makes it quick & easy to keep track of your data loggers. View real-time graphs & reports with a click of a button.
Get insights from your sensors on the fly, without having to login to your web portal.
Ontoto connect has diagnostic tools & integrations built-in. Easily troubleshoot device issues, export data & configure advanced settings all from the app.
Step 1: Attach the sensor to logger and turn on the logger
Step 2: Turn on the Application and click configure device. ( Note, your mobile should be within a 10m range of the logger )
Step 3: Click Scan, if the logger is powered on, the logger name will appear on the available device list with a default name “Ontoto”. Click connect, now connection will be made between logger and Android application via BLE (Bluetooth Low Energy Device )
Step 4: Now the Dashboard will be displayed
Step 5: Click “Run network test” to view the available network and 4G signal strength
Step 6: Click “Set Configuration” to set values such as reporting cycle ( every hour, every day, once a week … ) and logger name
Step 7: Click “Test Sensor” to ensure connected sensors are working properly
Step 8: Finally click “Start device” to bring the logger online and it will send an email according to the reporting cycle configuration. When the start button is pressed, the device will run the internal diagnostics and once the results are positive, the logger will go online. If there are any issues with the logger an alert message will be displayed.